With the Leica HDS , high-definition surveying has never been easier and friendlier for surveyors and measurement professionals. For example, it utilizes a . Leica HDS 3D Scanner. Supplied with: 2 Batteries; 2 Battery Chargers; Targets. The HDS requires Leica Cyclone (HDS is not supported . Find great deals for Leica ScanStation 3d Scanner HDS Shop with confidence on eBay!.
|Published (Last):||9 April 2016|
|PDF File Size:||9.55 Mb|
|ePub File Size:||15.94 Mb|
|Price:||Free* [*Free Regsitration Required]|
None Genuine Flight Case for Scanner. The surfaces of size mm x mm were placed on a plane on several desks made of chipboard and covered with white lamina, and measured under incidence angles of 0, 30, 50, 55, 75 and 90gon at distances of 15m and 25m.
Report item – opens in a new window or tab. Leica Disto Classic laser measure. This mainly depends on angle of incidence and physical characteristics of the hds30000, in particular reflectivity, absorbency and permeability.
The method assesses interior accuracy of the TLS system within a small field of view 1x1gon and therefore the results are not influenced by systematic errors such as collimation error and index error. The angle standard deviation specified by the manufacturer is 60 micro-radians, or, expressed in mgon, 2. Resume making your offerif the page does not update immediately. Peica Reading Kremen, T. Don’t miss the latest developments: Standard Shipping Standard Int’l Postage.
Leica disto d3a measuring tape Laser Kit Lot Tool swiss. Will usually ship within 1 business day of receiving cleared payment – opens in a new window or tab. The method is thus suited for scanners with standard deviation leiica direction from 1.
See all condition definitions – opens in a new window or tab Results This method can be used for all TLS systems emitting radiation in the visible part of the electromagnetic spectrum, as the digital camera is sensitive only to this radiation.
Learn More – opens in a new window or tab International shipping and import charges paid to Pitney Bowes Inc. The item you’ve selected was not added to your cart.
To the user the terrestrial laser scanner TLS is a black box; measurement procedure and data processing are treated as classified.
This means the item has been inspected, cleaned, and repaired to full working order and is in excellent condition.
Leica ScanStation 3d Scanner HDS | eBay
Supplied with Leica battery carry cases and non original padded scanner case. See other items More This amount is subject to change until you make payment.
Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc. Have one to sell? Distance Accuracy The quality of the leiac measured between TLS system and object depends on how much of the reflected signal will reach the TLS sensor. Please enter a valid ZIP Code.
Leica HDS3000 3D Scanner
Your country’s customs office can offer more details, or visit eBay’s elica on international trade. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab.
This item may or may not be in original packaging. Please enter 5 or 9 hrs3000 for the ZIP Code. Towards Optimal 3D Point Clouds. The results confirm that the most suitable are surfaces with diffuse reflection. You are covered by the eBay Money Back Guarantee if you receive an item that is not as described in the listing.
No additional import charges at delivery! Personal check, Other – See seller’s payment instructions, Credit card. Angle Accuracy There are few publications on the angle accuracy of TLS systems, and the methods described deal only with assessment of transverse error between measured and modelled object. Surfaces with specular reflection cause multiple reflections for steeper angles of incidence, resulting in hds3000 too long and thus incorrect.
One of the most used measurement principles for terrestrial laser scanners TLS is the spatial polar method: Medium-format Cameras for High-accuracy Mapping.
Refer to eBay Return policy for more details.
hds3000 This item will be shipped through the Global Shipping Program and includes international tracking. The method is based on a modified photogrammetric approach using direct linear transformation DLT in 2D. Test and Results Forty-six materials were selected on the basis of their reflectivity diagram, and tested. It is surprising that the accuracy of the HDS scanner is much better than specified by the vendor.
After transformation of the measurements leicw the TLS system into the plane system of the calibration field, errors are assessed as differences between photogrammetric coordinates and TLS coordinates.
Leica ScanStation 3d Scanner HDS 3000
An item that has been restored to working order by the eBay seller or a third party not approved by the manufacturer. These methods can be used only for relative accuracy assessment between different TLS systems. From this information and the coordinates of the position of the TLS, coordinates of points are calculated.
Back to home page. After switching off the lighting, the calibration field is measured with the TLS, during which procedure it is simultaneously photogrammetrically measured with an accuracy of approximately 0. The main types of reflection are diffuse and specular.